Install Tektronix Series Instrument Driver NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements. This is apart of the IEEE Place controls onto the Panel using the unplaced items box. Questions in this forum area are community supported. Delete the Yellow Error Wire.
|Date Added:||16 April 2004|
|File Size:||11.44 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Tektronix and Keithley
You can also troubleshoot and make trktronix function calls to your instrument to ensure that it is working as expected before moving into automation. With the File Name control, you can name the. Procedure Locate the communication ports on the back of the Tektronix Instrument. You can find these instrument drivers in the Instrument Driver Network.
NI has three labview instrument drivers for AFG3k. Back to Top 6. Albview add the ability to control the range of the measurement being taken. If you select the device, you can see additional properties, such as the device name, and any software or drivers that are installed on your system to support this device.
Should you require a time-sensitive answer, please contact your local Tektronix support center here. Your tektdonix type casting wrong. Click the Tektronix project to create a new copy.
Place the new control on the Panel by selecting the Unplaced items box and placing it on the Panel. What can I do?
Tektronix TDSB Digital Storage Oscilloscope – Instrument Driver – National Instruments
Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or selecting the Learning tekfronix from the top right. Tektronix and Keithley Web Forum Skip to content.
Google [Bot] and 1 guest. With the File Name control, you can name the new. If you are using a laptop, you likely have multiple USB ports and one Ethernet port.
Back to Top 2. Install Tektronix Series Instrument Driver NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements.
Place controls onto the Panel using fektronix unplaced items box. Finally, you can specify execution tfktronix, reporting, database logging, and connectivity to other enterprise systems for your test system. The Peak Amplitude Node reads in the waveform signal that the Read Multiple Waveforms Node creates and analyzes the data your Tektronix device collects.
Wire the system directory terminal on the top right of the Get System Directory Node to the base path terminal on the top left of the Build Path Node.
Customize the Panel of the VI to fit your applications needs. Rename the x and y axis of the histogram graph to correspond with the values you are measuring by clicking on the axis labels and replacing the text or by clicking labviee the axis labels and changing the Name text at the top of the Configuration Pane on the right.
These instruments also feature numerous triggering modes, deep onboard memory, and a driver software API that includes data streaming and analysis functions. Tektronix oscilloscopes have network capability to allow the computer to connect over this network.
Power on your Tektronix oscilloscope Instrument and allow time for initialization. Tektronix does not regularly laabview or guarantee responses in this area. Locate the communication ports on your host PC.
Rename this indicator “New File Path”. This is the Peak Amplitude measurement node that you are adding. They feature up to eight channels that can sample at speeds up to Place the unplaced controls and indicators on the Panel.