Error in imbmpinfo line 20 metadata. Can I manually set the sample rate on my InfiniiVision oscilloscope? Where can I find manuals for discontinued and obsolete Hewlett-Packard, Agilent and Keysight test equipment? Vinod Vinod view profile. After successful connection between my oscilloscope and Matlab, I runned the m file, but failed.
|Date Added:||3 November 2010|
|File Size:||9.70 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Yes, but with software version 5. This has been tested on an Agilent series oscilloscope.
How I can do it? Could anyone give me some suggestion to resolve this problem? Updated 01 Sep Ideal for engineers who need differential measurements up to 30V rms floating, and with the BND power option. You need to upgrade your oscilloscope to the latest msi6104a Error in imbmpinfo line 20 metadata. I’m building a complete matlab GUI suite for fast testing, but I’m unable to capture data from the digital analyzer.
Overall i get these messages: Most Keysight products have PDF manuals that can be found on product pages. It eventually does output a square-wave to figure 1 I’m using the Demo 2 system with the oscilloscope but the output is of the wrong sign I expect a square between 0 and 2. Guillermo L Guillermo L view profile. MATLAB extends the functionality of Agilent oscilloscopes by enabling you to analyze and visualize live or acquired oscilloscope signals, make application-specific measurements, create and apply your own filters and equalization methods, and automate tests.
The user can customize the code to allow for different segment number of points, sample rates, etc. Compact and versatile enough for field work the entire day. Although the sample rate cannot be directly set on any InfiniiVision oscilloscope, it can be set indirectly by adjusting the oscilloscopes time per division setting.
Agilent 6000 Series Oscilloscope
Select the China site in Chinese or English for best site performance. Tags Add Tags keysight not a function.
Select a Web Site Choose a web site to get translated content where available and see local events and offers. Like most other digitizing oscilloscopes, the new Series is not designed to offer floating measurement by itself.
Message 4 of 5. This example shows you how you can use MATLAB to control an Agilent oscilloscope, capture a waveform and grab a screenshot on the instrument.
Again – modify the parameter to a value supported by the scope.
Keysight | Electronic Measurement Products | Agilent
Does this work with Agilent Infiniium oscilloscopes? The error meassage is Warning: Other MathWorks country sites are not optimized for visits from your location. I am using the data acquisition tool box for the 1st time and also the code is new to me. Within the “Save” menu, press Settings and select the max number of points.
Select a Web Site
For more information on the SCPI commands used to control the instrument, refer to your instrument’s programmer’s guide. Florian Jungbold Florian Jungbold view profile. I’m writing code of similar program and I want make capture of signal’s realization, during the period of time.
What is the difference between labvview Auto and Normal trigger modes on InfiniiVision oscilloscopes?
Modify this example accordingly if your instrument screen displays errors as you step through the code. In auto trigger mode, the trigger will be forced if the specified conditions are not met. Does the following code also work for agilent infiniium D MSO?